- Capacity up to 6 Kg of sample.
- Three-dimensional movement.
- It can fit wet and dry sieves.
- It is controlled by a microprocessor.

The analytic **sieve shaker OASS203** is designed to obtain reproducible results in accordance with the standard ISO 9001 for measuring and control equipment. It is an essential device for research laboratorios and for quality assessment of any type of industries during the analysis of the production process. It allows to define mechanic characteristics of particles, concentration by joining forces, miscibility, perfomance with regard to stress, organoleptic characteristics, etc.

- Capacity up to 6 Kg of sample.
- Three-dimensional movement.
- It can fit wet and dry sieves.
- It is controlled by a microprocessor.

- Standard lock system easy to program provided with the sieve.
- Adjustment of the sieve power (100% corresponds to 6400 RPM). This allows better spread of the simple through the sieve and better efficiency in the sieve process.
- It is programable up to 16 memories. Time can be adjusted from 10 seconds to 99 minutes and hold position.
- Adjustable by intervals from 1 to 99 seconds.

- Extremely silent. It has the leasr noise level of those available.
- Metal cover. It is tough and stable.
- Electric protection with ground power and uses.

- Stainless steel sieves AISI 316 for chain mails and AISI 304 made of perforated plate with sealing gasket which is market with indelible lase.
- Sieves diameter: 203 mm (8”), 200 mm.
- Capacity uo to 8 of 50 mm height or 16 of 25 mm, total width of the fall 400 mm.
- Range of particle sizes which can by analyzed: from 20 µ to 125 mm .
- The calibration certificate is available.

Reference |
External dimensions (w x d x h) |
Power Consumption (w) | Voltage (V) | Frecuency (Hz) | Weight (Kg) | ||

TA 005 | 280 | 370 | 765 | 120 | 220-240 | 50-60 | 14,5 |

TA 006 | 280 | 370 | 765 | 120 | 110-120 | 50-60 | 14,5 |

Available dimensions: 200 mm Ø x 50 mm (h), 200 mm Ø x 25 mm (h), 203 ( 8”) mm Ø x 50 mm (h),

203 mm ( 8”) Ø x 25 mm (h).

Stainless steel AISI 316 for chain mails sieves and AISI 304 for those made of perorated plate.

Sieves made in stainless steel 200 x 50 mm Sieves made in stainless steel 8" (203 x 50 mm)Standard ISO 3310-2: Perforated plates | Standard ISO 3310-1: Metalic mesh | |||

Ref | Mesh (mm) | Ref | Mesh(mm) | |

PI 065 | 125.00 | PI 079 | 8.00 | |

PI 069 | 100.00 | PI 080 | 6.30 | |

PI 070 | 80.00 | PI 081 | 5.00 | |

PI 071 | 63.00 | PI 082 | 4.00 | |

PI 072 | 50.00 | PI 297 | 3.15 | |

PI 073 | 40.00 | PI 083 | 2.50 | |

PI 074 | 25.00 | PI 348 | 2.36 | |

PI 075 | 20.00 | PI 084 | 2.00 | |

PI 076 | 16.00 | PI 321 | 1.7 | |

PI 077 | 12.50 | PI 085 | 1.6 | |

PI 078 | 10.00 | PI 086 | 1.25 | |

PI 087 | 1.00 | |||

PI 088 | 0.80 | |||

PI 089 | 0.63 | |||

PI 090 | 0.50 | |||

PI 091 | 0.40 | |||

PI 146 | 0.315 | |||

PI 092 | 0.25 | |||

PI 093 | 0.20 | |||

PI 094 | 0.16 | |||

PI 095 | 0.125 | |||

PI 096 | 0.100 | |||

PI 097 | 0.080 | |||

PI 098 | 0.063 | |||

PI 099 | 0.050 | |||

PI 100 | 0.040 | |||

PI 066 | Receiver | |||

PI 067 | Cover | |||

PI 351 | Receiver wet system | |||

PI 350 | Covert wet system |

Standard ASTM E323: Perforated plates | Standard ASTM E11: Metalic mesh | |||

Ref | Mesh | Ref | Mesh | |

PI 150 | 5" | PI 169 | 3/8" | |

PI 151 | 4,24" | PI 170 | 5/16" | |

PI 152 | 4" | PI 171 | 0.265" | |

PI 153 | 3 1/2" | PI 172 | 1/4" | |

PI 154 | 3" | PI 173 | 3 1/2" | |

PI 155 | 2 1/2" | PI 174 | No. 4 | |

PI 156 | 2.12" | PI 175 | No. 5 | |

PI 157 | 2" | PI 176 | No. 6 | |

PI 158 | 1 3/4" | PI 177 | No. 7 | |

PI 159 | 1 1/2" | PI 178 | No. 8 | |

PI 160 | 1 1/4" | PI 179 | No. 10 | |

PI 161 | 1.06" | PI 180 | No. 12 | |

PI 162 | 1" | PI 182 | No. 16 | |

PI 163 | 7/8" | PI 181 | No.14 | |

PI 164 | 3/4" | PI 183 | No. 18 | |

PI 165 | 5/8" | PI 184 | No. 20 | |

PI 166 | 0.53" | PI 185 | No. 25 | |

PI 167 | 1/2" | PI 186 | No.30 | |

PI 168 | 7/16" | PI 187 | No. 35 | |

PI 188 | No. 40 | |||

PI 189 | No. 45 | |||

PI 190 | No. 50 | |||

PI 191 | No. 60 | |||

PI 192 | No. 70 | |||

PI 193 | No. 80 | |||

PI 194 | No. 100 | |||

PI 195 | No. 120 | |||

PI 196 | No. 140 | |||

PI 197 | No. 170 | |||

PI 198 | No. 200 | |||

PI 199 | No. 230 | |||

PI 200 | No. 270 | |||

PI 201 | No. 325 | |||

PI 250 | No. 400 | |||

PI 202 | Cover | |||

PI 203 | Receiver | |||

PI 221 | Cover wet system | |||

PI 235 | Receiver wet system |